Doppler sonography/physical principle

Introduction
Sonongraphy, also sometimes called ultrasound scanning is a safe, painless and non-invasive imaging technique which involves the use of high frequency sound waves to produce real time images of internal organs, tissues and bloodflow in the body. During this process, high frequency sound waves of between 1-5 MHz are transmitted into the body using a transducer, the sound waves are reflected by the tissues being examined back to the same transducer, which then relays the message to the ultrasound machine. The machine is able to interprete this data by calculating the distance between the transducer and the tissues using the speed of sound in soft tissue (1,540 m/s) and the time taken for the echo to return; this enables a two-dimensional image to be formed.

The Doppler-Sonography is a special type of Sonography that enables the physician to display the the bloodflow in blood vessels and the heart in acoustic and visual qualities using the Doppler effect of soundwaves. The Doppler effect is simply the observed change in frequency when relative motion exists between the source of a sound wave and the observer. For example, the change in pitch when an ambulance zooms past a pedestrian on the street (For further explanation, see article on Doppler effect). In Doppler Sonography it is still necessary to use the method of usual Sonography to find the target structure. The target structures of Doppler Sonography need to be vessels or the heart since a moving object or fluid (in this case blood) is required to create the Doppler effect. Common examined organs or tissues are for instance the common carotid artery, the heart and also the vessles of the brain. Nowadays, modern medical ultrasound device can use Doppler Sonography in order to evaluate the bloodflow of the visualized structures. Because Doppler-Sonography is fast, relatively cheap, and non-invasive it has become essential in the diagnosis of vessels in various areas of medicine such as internal medicine, gynaecology, angiology and cardiology. In this part of the article we are going to discuss in detail how the Doppler effect is applied to the medical ultrasound device.

Applied Doppler effect in Sonography
In Sonography, the Doppler effect works slightly different compared to the example provided above as the blood does not emmit any sound but rather reflects the soundwaves produced by the transducer. Through the reflection of the soundwaves by the erythroctes and their movement due to bloodflow the blood behaves like a moving source of sound. Due to the fact that echoes from stationary tissues undergo little to no changes in frequency, the only detectable changes are those produced by the flow of blood as they undergo the Doppler effect. These differences are measured in terms of a phase shift (Doppler shift), and it is thus possible to obtain a Doppler frequency. These can then be further processed to produce a colour flow display or spectrum.



The above diagram shows the formula on which the entire principle of Doppler Sonography is based. It shows that the Doppler shift is determined by four main factors:


 * The frequency of the transmitted beam (ft) - The higher the frequency of the ultrasound  beam transmitted by the transducer to the tissues, the greater the value obtained for doppler shift. In Doppler sonography, a compromise is usually made concerning the ultrasound frequency transmitted because despite the fact that they produce a greater doppler shift, higher ultrasound frequencies have a lower penetrating power.


 * The velocity of blood (V)- The doppler shift is also directly proportional to the velocity of blood; a quicker velocity means a correspondingly larger value for the doppler shift.


 * The speed of sound in soft tissues (c)- The formula shows that the doppler shift is indirectly proportional to the rate at which sound travels in soft body tissues. This value is usually constant at 1,540 m/s.


 * The angle of incidence between the ultrasound beam and the direction of flow. (θ)- This is a major factor that determines the strength of the Doppler signal. It is extensively explained in the figure below.

Dependence of Doppler Shift on beam-flow angle (Angle of insidence)


When a physician is trying to examine a vessel via Doppler Sonography it is of utmost importance to take the beam-flow angle into consideration as the Doppler effect is dependant on the position of the reseiver to the source of sound. As shown in the picture with the ambulance we obtain either a much higher frequency when movement of the source of sound is towars the receiver or a much lower frequency when the movement of the source of sound is away from the receiver. Therefore the optimal value for the Doppler effect would be at 180° or 0° to the source of sound. Now imagine the observer of the ambulance is standing at 90° to the driving ambulance. The waves the reseiver is hearing now are neither of a higher nor a lower frequency which means there is no doppler effect. This phenomena can be seen even clearer when measureing the distances between the waves at 90° and comparing them to the the distance between the waves when the ambulance is not moving. You should obtain approximatley the same frequency. Therefore the sound the person will hear at 90° to the moving ambulance will be the same sound as if the ambulance was not in movement. This relation is expressed in the formula through cos θ. The cosine of 0° and 180° is 1 and -1. The cosine of 90° is 0. That is why the optimal angle of insidence or beam-flow angle is fixed between 0°- 60° or 120°-180°. In both cases the Doppler signal is high enough to measure a accurate value of the velocity of blood. Trough the angle of inscidence the device is able to display different directions of blood-flow as opposite directions of flow will have the same magnitude but either a positive (0°-60°) or a negative (120°-180°) value as the angle is inverted with changing direction. Variying among ultrsound devices blood flowing towards the probe it coloured differently to blood flowing away from the probe. That is an additional method to differentiate between arterys and veins.

Effect of the Doppler angle in the sonogram. (A) higher-frequency Doppler signal is obtained if the beam is aligned more to the direction of flow. In the diagram, beam (A) is more alilgned than (B) and produces higher-frequency Doppler signals. The beam/flow angle at (C) is almost 90° and there is a very poor Doppler signal. The flow at (D) is away from the beam and there is a negative signal.